C16/D16 Flow Indicator
Flow Indicator with tubing adapter for the C16 and D16. (03-0107)
Flow Meter (0-1 SCFH) with tubing adapter for the C16/D16 PortaSense detectors. This flow meter is used to verify the flow rate. To use it, turn on the C16/D16 and connect the sampling wand. Push the flow meter’s tubing adapter onto the wand, and hold the meter vertical. Verify the flow rate exceeds 0.5 SCFH (250 cc/min). If the flow rate cannot be achieved, inspect the wand and manifold for blockages.
|Call For Price||No|
|Monitor Type||C16, D16|