Hydrogen Sulfide Puffer Test Gas Generator
Bump Tester for Hydrogen Sulfide
The Puffer is a portable instrument that can be used to produce a small amount of sample gas on site to verify the operability of the your gas monitor. The puffer contains a small hydrogen sulfide gas generator and then a simple hand pump is used use to move the gas through the tube to deliver hydrogen sulfide gas to the gas monitor that you need to test.
HOW TO USE
1. Push the activation switch and hold until the % life indicator flashes 3 times.
2. Release the switch and wait until the indicator stops flashing (about 45 seconds).
3. Once it stops flashing, the sample gas is ready to be idspensed.
a. push the black rubber bladder on the front of the unit to propel the gas through the tube towards the sensor that needs to be tested.
b. For monitors with internal pump, the plug next to the outlet tube can be removed (see image below) and connect the sample wand to the exposed port. The black rubber bladder does NOT need to be pushed. The gas detector's pump will draw air from the puffer.
|Call For Price||No|
|Monitor Type||B12, C16, D12, D12-IR, F12, S-200, S-300, S-500|