K-5003 Hydrazine Vacu-Vials Kit is used to determine Hydrazine levels from 0-1.20 ppm.
This kit requires the use of a CHEMetrics Direct-Readout Photometer or a spectrophotometer capable of accepting a 13 mm diameter round cell. Instrument sold separately.
Why Choose Vacu-vials?
CHEMetrics Vacu-vials self-filling ampoules are a great way to test for dissolved substances in liquid. Unlike our visual test kits, the Vacu-vials test kit allows you to obtain an instrumental result using the CHEMetrics Direct-Redout Photometers or any spectrophotometer capable of accepting a 13m diameter round cell. When using equipment from another manufacturer, set the spectrophotometer to absorbance mode, select the wavelength specified by the Vacu-vials kit instructions, and follow the test procedure. K-5003 Hydrazine Vacu-Vials Kit is used to determine Hydrazine levels from 0-1.20 ppm.
Safe and Reliable Results
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Vacuvials test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
CHEMetrics’ hydrazine test kits employ the PDMAB, paradimethylaminobenzaldehyde chemistry. PDMAB in acid solution reacts with hydrazine to form a yellow product. Results are expressed as ppb (ug/L) or ppm (mg/L) N2H4.
Hydrazine is a powerful reducing agent that is used in various chemical processes and in boiler water as an oxygen scavenger. To control corrosion, residual hydrazine typically is maintained in the 0.05 to 0.1 mg/L range. Higher levels may be used to guard against corrosion when the boiler is out of service for an extended period.
How to use:
|1. Fill the sample cup to the 25 mL mark with the sample to be tested.||2. Place the Vacu-vial ampoule tip-first into the sample cup. Snap the tip and the ampoule will automatically fill.|
|3.To mix the ampoule, tilt it from side to side, allowing the bubble to travel from end to end.||4. Dry the ampoule|
|5. Wait 10 minutes for color development.||6. Insert the Vacu-vial ampoule into the photometer, flat end first, and obtain a reading in ppm.|
WARNING! This product can expose you to chemicals including mercury, which is known to the State of California to cause birth defects or other reproductive harm. For more information go to www.P65Warnings.ca.gov.
More CHEMetrics Kits for Hydrazine
|K5005D||PDMAB||0.25 ppm||0-0.25 ppm||$52.00|
|K5005A||PDMAB||0.5 ppm||0-0.5 ppm||$52.00|
|K5005B||PDMAB||1 ppm||0-50 ppm||$52.00|
|K5005C||PDMAB||10 ppm||0-500 ppm||$54.00|
Downloads and Links:
|Call For Price||No|
|4-20 mA Output||No|
|Power Requirements||4 AAA Batteries|