K-5510 Hydrogen Peroxide
Why Choose CHEMetrics Chemets kits?
CHEMetrics Chemets self-filing reagent ampoules are a great way to test for dissolved substances in liquids. These visual test kits allow you to obtain results by measuring the color of the sample according to the color comparators provided with the kit. This unique method allows you to attain results quickly and accurately with minimal contact with potentially hazardous materials.
Safe and Reliable Results:
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Chemets test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
-30 self-filling Ampoules
-low and high-range comparators
-1 25 mL Sample Cup
The K-5510 Hydrogen Peroxide Kit was designed to accurately measure the levels of Hydrogen Peroxide in water from 0-0.8 ppm and 1-10 ppm. K-5510 uses the Ferric Thiocyanate Method, which consists of ammonium thiocyanate and ferrous iron in acid solution. Hydrogen peroxide oxidizes ferrous iron to the ferric state, resulting in the formation of a red thiocyanate complex. Chlorine will not interfere with this method. Ferric iron and peracetic acid will interfere. Results are expressed as ppm (mg/L) H2O2.
|1. Fill the sample cup to the 25 mL mark with the sample to be tested.
|2. Place the ampoule tip-first into the sample cup. Snap the tip and the ampoule will automatically fill.
|3. To mix the ampoule, tilt it from side to side, allowing the bubble to travel from end to end.
|4. Dry the ampoule. Test results should be obtained between 30 seconds and 1 minute of snapping the tip.
|5. Place the ampoule into the center of the round comparator and hold it toward a source of light, viewing from the bottom. Rotate the comparator until the color below the ampoule shows the closest match.
|6. For higher concentrations, use flat comparator. Place the ampoule between the color standards, moving it along the comparator until you can determine the closest match.
For more detailed instructions, see the link at the bottom of the page.
WARNING! This product can expose you to chemicals including methanol, which is known to the State of California to cause birth defects or other reproductive harm. For more information go to www.P65Warnings.ca.gov.
More CHEMetrics Kits for Hydrogen Peroxide:
|Ceric Sulfate Tirant with Ferroin Indicator
|0.1-1.0% (up to 20% with dilution)
|0-25 & 30-300 ppm
|0-50 & 60-600 ppm
|0-100 & 120-1,200 ppm
|0-1000 & 1200-12,000 ppm
|0-0.8 & 1-10 ppm
|Quickbooks Stock Level
|Call For Price
|Hydrogen Peroxide (H2O2)
|4-20 mA Output
|4 AAA Batteries
|0-0.8 ppm and 1-10 ppm