K-9003 Silica Vacu-vials Kit was designed to obtain Silica levels from 0-10.00 ppm
- 30 Ampoules
- 25 mL Sample Cup and Top
- Activator Solution
- Neutralizer Solution
- Ampoule Blank
- Instruction Manual
- Put the Zero ampoule into the device and cover with the light shield piece to zero out the unit. (NOTE: This must be zeroed any time the unit is turned off and turned back on)
- Fill to the 15 mL mark in the sample cup with sample water.
- Add 10 drops of Activator Solution, cap Sample Cup, and shake gently. Wait 4 minutes
- Add 5 drops of Neutralizer Solution, cap Sample Cup, and shake gently. Wait 1 minute.
- Snap the end of the Vacu-vial in the water on the bottom/side of the 25 mL sample cup.
- Mix the contents of the ampoule by tipping back and forth a few times.
- Dry the vacu-vial and wait 2 minutes for color development.
- Place the vacu-vial ampoule into the device (flat end first) for readout.
|Call For Price||No|
|4-20 mA Output||No|
|Power Requirements||4 AAA Batteries|
Write Your Own Review