K-5530 Hydrogen Peroxide
K-5530 Peroxide Titrets Kit was designed to obtain Peroxide levels in 0.1-1.0% (up to 20% with dilution)
Why Choose CHEMetrics Titrets?
CHEMetrics Titrets self-filing reagent ampoules make it easy to perform quick, accurate titrations of water samples. The handheld titrettor that comes with the kit allows the user to easily and reliably control the flow of the solution being measured without the need to count drops or use burets or other equipment. This unique design allows you to perform a titration anywhere and achieve reliable results for a variety of substances.
Safe and Reliable Results:
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Tritrets test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
-30 self-filling Ampoules with valve assemblies
-1.0 mL syringe
-3.0 mL syringe
-1 25 mL Sample Cup
The K-5530 Hydrogen Peroxide Kit was designed to accurately measure the levels of Hydrogen Peroxide in water from 0.1-1.0% (up to 20% with dilution). K-5530 uses the Ceric Sulfate Titrimetric Method. CHEMetrics developed a titrimetric method using ceric sulfate as the titrant and ferroin as the endpoint indicator. A color change from green to orange signals the end of the titration. Results are expressed as percent (%) H2O2. The test range can be modified by performing a sample dilution. Details are provided in the kit instructions for ranges of 0.01 - 0.1% through 2- 20%.
How to Use:
Note: Images used in the following table show a total hardness and colors may differ from the ones shown. Read written instructions to know what color changes to expect from your substance.
|1. Using the syringe provided, obtain 2 mL of the sample to be tested and dispense it into the empty sample cup.||2. Dilute the contents of the sample cup to the 20 mL mark with distilled water.|
|3. Snap the tip of the ampoule at the black snap ring.||4. Lift the control bar and insert the Titret assembly into the Titrettor.|
|5. Hold the Titrettor with the sample pipe in the sample. Press the control bar firmly, but briefly, to pull in a small amount of sample. The contents will turn green.||6. Press the control bar again to draw another small amount of sample into the ampoule.|
|7. Rock the entire assembly to mix the contents of the ampoule. Watch for a color change from blue to colorless. Repeat steps 6 and 7 until a permanent color change occurs.||8. When the color of the liquid in the ampoule changes through blue to brownish-orange, remove the ampoule from the Titrettor. Hold the ampoule tip pointed upward and read the scale opposite the liquid level. Multiply scale unit by 0.1 for test results in % hydrogen peroxide (H2O2).|
More CHEMetrics Kits for Hydrogen Peroxide:
|K5530||Ceric Sulfate Tirant with Ferroin Indicator||0.10%||0.1-1.0% (up to 20% with dilution)||$33.00|
|K5510D||Ferric Thiocyanate||5 ppm||0-25 & 30-300 ppm||$71.00|
|K5510A||Ferric Thiocyanate||10 ppm||0-50 & 60-600 ppm||$71.00|
|K5510B||Ferric Thiocyanate||20 ppm||0-100 & 120-1,200 ppm||$71.00|
|K5510C||Ferric Thiocyanate||200 ppm||0-1000 & 1200-12,000 ppm||$72.00|
|K5510||Ferric Thiocyanate||0.05 ppm||0-0.8 & 1-10 ppm||$66.00|
|K5543||Ferric Thiocyanate||0-6.00 ppm||$30.00|
|Call For Price||No|
|Gas||Hydrogen Peroxide (H2O2)|
|4-20 mA Output||No|
|Power Requirements||4 AAA Batteries|