Why Choose CHEMetrics Vacuettes kits?
VACUettes ampoules are designed for highly concentrated samples. They employ a patented auto-dilution feature that eliminates the need for a time-consuming and error-prone preliminary dilution. As a result, the entire test takes only 2 to 3 minutes, with a rate of accuracy comparable to a volumetric procedure. The basic design of these 7 mm ampoules is the same as CHEMets ampoules, however, a capillary tip is attached to the tip of each ampoule.
Safe and Reliable Results:
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Vacuettes test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
-30 self-filling Ampoules
-25mL sample cup
-200 uL minipet
The K-2504B CHEMets Visual Kit is a visual indicator kit for measuring chlorine in water levels from 0-100 and 0-500ppm. This device uses the DPD Method for testing. Free chlorine oxidizes DPD (N,N-diethyl-p-phenylenediamine) to form a pink colored species in direct proportion to the chlorine concentration. Total chlorine, the sum of free and combined chlorine, is determined by adding an excess of potassium iodide to the sample. Chloramines (combined chlorine) oxidize the iodide to iodine. The iodine then oxidizes DPD to the pink colored species. Other halogens, ozone, and halogenating agents will produce high test results. Chlorine at concentrations significantly above the test range may prevent proper color development, causing low test results.
|1. Place the yellow pipette tip firmly onto the end of the minipet.||2. Depress the plunger on the minipet.
Immerse the tip in the sample to be tested
and release the plunger. A portion of the
sample will be drawn into the tip.
|3. Hold the minipet over the sample cup, and
depress the plunger to dispense sample.
|4. Dilute the contents of the sample cup to
the 20 mL mark with distilled water.
|5. Place the ampoule, tip first, into the sample cup. Snap the tip. The ampoule will fill automatically.||6. To mix the ampoule, invert it several times,
allowing the bubble to travel from end to end. Dry the ampoule. Obtain the test result 1 minute after snapping the tip.
|7. Place the ampoule into the center of the round comparator and hold it toward a source of light, viewing from the bottom. Rotate the comparator until the color below the ampoule shows the closest match.||8. For higher concentrations, use flat comparator. Place the ampoule between the color standards, moving it along the comparator until you can determine the closest match.|
For more detailed instructions, see the link at the bottom of the page.
More CHEMetrics Kits for Chlorine:
|K2513||DPD (free and total)||0-5.0ppm|
|K2511||DDPD (free & total)||0.04 ppm||0-0.20 ppm|
|K2504||DPD (free & total)||0.05 ppm||0-1 & 0-5 ppm|
|K2504D||DPD (free & total)||2.5 ppm||0-25 & 0-125 ppm|
|K2504A||DPD (free & total)||5 ppm||0-50 & 0-250 ppm|
|K2504B||DPD (free & total)||10 ppm||0-100 & 0-500 ppm|
|K2504C||DPD (free & total)||40 ppm||0-400 & 0-2000 ppm|
|Quickbooks Stock Level||0.000000|
|Call For Price||No|
|4-20 mA Output||No|
|Range||0-100ppm & 0-500ppm|