K-7913 Peracetic Acid
Why Choose Vacu-vials?
CHEMetrics Vacu-vials self-filling ampoules are a great way to test for dissolved substances in liquid. Unlike our visual test kits, the Vacu-vials test kit allows you to obtain an instrumental result using the CHEMetrics Direct-Redout Photometers or any spectrophotometer capable of accepting a 13m diameter round cell. When using equipment from another manufacturer, set the spectrophotometer to absorbance mode, select the wavelength specified by the Vacu-vials kit instructions, and follow the test procedure. The K-7913 Peracetic Acid Kit was designed to accurately measure the levels of Peracetic Acid in water from 0-5.0 ppm.
Safe and Reliable Results
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Vacuvials test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
The K-7913 uses the DPD Method, in which the sample is treated with an excess of potassium iodide. Peracetic acid oxidizes the iodide to iodine. The iodine then oxidizes DPD (N, N-diethyl-p-phenylenediamine) to form a pink-colored species that is directly proportional to the peracetic acid concentration in the sample. Results are expressed as ppm (mg/L) peracetic acid.
Various oxidizing agents such as halogens, ozone, and cupric ions will produce high test results. Hydrogen peroxide does not interfere if present at levels comparable to the peracetic acid levels.
How to use:
|1. Add 5 drops of Activator Solution to the empty sample cup.||2. Fill the sample cup to the 25 mL mark with the sample to be tested.|
|3. Immediately place the Vacu-vial ampoule tip-first into the sample cup. Snap the tip and the ampoule will automatically fill.||4. To mix the ampoule, tilt it from side to side, allowing the bubble to travel from end to end.|
|5. Dry the ampoule and wait 1 minute for color development.||6. Insert the Vacu-vial ampoule into the photometer, flat end first, and obtain a reading in ppm.|
More CHEMetrics Kits for Peracetic Acid
|K7904||DPD||0.05ppm||0-1 & 0-5 ppm||$66.00|
|K7904A||DPD||7 ppm||0-70 & 0-300 ppm||$71.00|
|K7904B||DPD||15 ppm||0-130 & 0-600 ppm||$71.00|
|K7904C||DPD||120 ppm||0-1200 & 0-6000 ppm||$72.00|
|K7909D||DPD||3 ppm||0-30 & 0-150 ppm||$71.00|
|Call For Price||No|
|4-20 mA Output||No|
|Power Requirements||4 AAA Batteries|