Why Choose CHEMetrics Vacuettes kits?
VACUettes ampoules are designed for highly concentrated samples. They employ a patented auto-dilution feature that eliminates the need for a time-consuming and error-prone preliminary dilution. As a result, the entire test takes only 2 to 3 minutes, with a rate of accuracy comparable to a volumetric procedure. The basic design of these 7 mm ampoules is the same as CHEMets ampoules, however, a capillary tip is attached to the tip of each ampoule.
Safe and Reliable Results:
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Vacuettes test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
-30 self-filling Ampoules
-Diluter snapper cup
-Sample cup top
-Micro test tube
The K-9510A Sulfide Kit was designed to accurately measure the levels of Sulfide in water from 0-60 ppm and 60-600 ppm. K-9510 A uses the Methylene Blue Method. CHEMetrics test kits measure total acid-soluble sulfides and employ the methylene blue methodology. Sulfides react with dimethyl-p-phenylenediamine in the presence of ferric chloride to produce methylene blue. Results are expressed as ppm (mg/L) S.
|1. Fill the diluter snapper cup to the -ml- mark with distilled water. Add 4 drops of Activator Solution. Cap the cup and shake to mix the contents.||2. Fill the micro-test tube approximately halfway with the sample to be tested.|
|3. Holding the VACUette almost horizontally,
touch the tip to the contents of the micro-test
|4. Place the VACUette between the vertical tip
guides on the inside of the dilutor snapper
cup and snap the ampoule tip.
|5.To mix the ampoule, invert it several times,
allowing the bubble to travel from end to end.
|6. Dry the ampoule and wait 5 minutes for color development.|
|7. Place the ampoule into the center of the round comparator and hold it toward a source of light, viewing from the bottom. Rotate the comparator until the color below the ampoule shows the closest match.||8. For higher concentrations, use flat comparator. Place the ampoule between the color standards, moving it along the comparator until you can determine the closest match.|
For more detailed instructions, see the link at the bottom of the page.
More CHEMetrics Kits for Sulfide:
|K9503||Methylene Blue||0-3.00 ppm or 0-10.00 ppm|
|K9523||Methylene Blue||0-6.00 ppm|
|K9510||Methylene Blue||0.05 ppm||0-1 & 1-10 ppm|
|K9510D||Methylene Blue||5 ppm||0-30 & 30-300 ppm|
|K9510A||Methylene Blue||10 ppm||0-60 & 60-600 ppm|
|K9510B||Methylene Blue||20 ppm||0-120 & 120-1200 ppm|
|K9510C||Methylene Blue||200 ppm||0-1200 & 1200-12,000 ppm|
|Call For Price||No|
|4-20 mA Output||No|
|Power Requirements||4 AAA Batteries|
|Range||0-60 ppm and 60-600 ppm|