K-9503 Sulfide Vacu-Vials Kit is used to determine Sulfide levels from 0-3.00 ppm.
This kit requires the use of a CHEMetrics Direct-Readout Photometer or a spectrophotometer capable of accepting a 13 mm diameter round cell. Instrument sold separately.
Why Choose Vacu-vials?
CHEMetrics Vacu-vials self-filling ampoules are a great way to test for dissolved substances in liquid. Unlike our visual test kits, the Vacu-vials test kit allows you to obtain an instrumental result using the CHEMetrics Direct-Redout Photometers or any spectrophotometer capable of accepting a 13m diameter round cell. When using equipment from another manufacturer, set the spectrophotometer to absorbance mode, select the wavelength specified by the Vacu-vials kit instructions, and follow the test procedure. The K-9503 Sulfide Kit was designed to accurately measure the levels of Sulfide in water from 0-3.00 ppm.
Safe and Reliable Results:
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Vacuvials test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
-30 self-filling Ampoules
-25 mL Sample Cup
K-9503 uses the Methylene Blue Method. CHEMetrics test kits measure total acid-soluble sulfides and employ the methylene blue methodology. Sulfides react with dimethyl-p-phenylenediamine in the presence of ferric chloride to produce methylene blue. Results are expressed as ppm (mg/L) S.
How to Use:
|1. Fill the sample cup to the 25 mL mark with the sample to be tested.||2. Add 5 drops of the Activator Solution. Stir to mix the contents of the cup.|
|3. Immediately place the Vacu-vial ampoule tip-first into the sample cup. Snap the tip and the ampoule will automatically fill.||4. To mix the ampoule, tilt it from side to side, allowing the bubble to travel from end to end.|
|5. Dry the ampoule and wait 5 minutes for color development.||6. Insert the Vacu-vial ampoule into the photometer, flat end first, and obtain a reading in ppm.|
More CHEMetrics Kits for Sulfide:
|K9510||Methylene Blue||0.05 ppm||0-1 & 1-10ppm||$66.00|
|K9510D||Methylene Blue||5 ppm||0-30 & 30-300 ppm||$71.00|
|K9510A||Methylene Blue||10 ppm||0-60 & 60-600 ppm||$71.00|
|K9510B||Methylene Blue||20 ppm||0-120 & 120-1200 ppm||$71.00|
|K9510C||Methylene Blue||200 ppm||0-1200 & 1200-12,000 ppm||$72.00|
|Call For Price||No|
|4-20 mA Output||No|
|Power Requirements||4 AAA Batteries|
|Range||0-3.00 ppm or 0-1.00 ppm depending on reader|