Why Choose CHEMetrics Chemets kits?
CHEMetrics Chemets self-filling reagent ampoules are a great way to test for dissolved substances in liquids. These visual test kits allow you to obtain results by measuring the color of the sample according to the color comparators provided with the kit. This unique method allows you to attain results quickly and accurately with minimal contact with potentially hazardous materials.
Safe and Reliable Results:
While traditional methods often require multiple steps and thorough preparation, these test kits allow you to obtain dependable results with a few simple steps. By simplifying the testing method, Chemets test kits reduce the potential for operator error. Additionally, each of the ampoules contained in the test kits is vacuum sealed which helps eliminate the problem of obtaining inaccurate results from stale or unstable reagents. These reagents are stored right in the tip of each ampoule and sealed in glass, thereby minimizing direct contact with the chemicals.
-30 self-filling Ampoules
-low and high-range comparators
-1 25 mL Sample Cup
-1 mL syringe
The K-2504 CHEMets Visual Kit is visual indicator kit for measuring chlorine in water levels from 0-1 and 0-5ppm. This device uses the DPD Method for testing. Free chlorine oxidizes DPD (N,N-diethyl-p-phenylenediamine) to form a pink colored species in direct proportion to the chlorine concentration. Total chlorine, the sum of free and combined chlorine, is determined by adding an excess of potassium iodide to the sample. Chloramines (combined chlorine) oxidize the iodide to iodine. The iodine then oxidizes DPD to the pink colored species. Other halogens, ozone and halogenating agents will produce high test results. Chlorine at concentrations significantly above the test range may prevent proper color development, causing low test results.
|1. Fill the sample cup to the 25 mL mark with the sample to be tested.||2. Place the ampoule tip-first into the sample cup. Snap the tip and the ampoule will automatically fill.|
|3. To mix the ampoule, tilt it from side to side, allowing the bubble to travel from end to end.||4. Dry the ampoule and wait 1 minute for color development.|
|5. Place the ampoule into the center of the round comparator and hold it toward a source of light, viewing from the bottom. Rotate the comparator until the color below the ampoule shows the closest match.||6. For higher concentrations, use flat comparator. Place the ampoule between the color standards, moving it along the comparator until you can determine the closest match.|
For more detailed instructions, see the link at the bottom of the page.
More CHEMetrics Kits for Chlorine:
|K2513||DPD (free and total)||0-5.0ppm|
|K2511||DDPD (free & total)||0.04 ppm||0-0.20 ppm|
|K2504||DPD (free & total)||0.05 ppm||0-1 & 0-5 ppm|
|K2504D||DPD (free & total)||2.5 ppm||0-25 & 0-125 ppm|
|K2504A||DPD (free & total)||5 ppm||0-50 & 0-250 ppm|
|K2504B||DPD (free & total)||10 ppm||0-100 & 0-500 ppm|
|K2504C||DPD (free & total)||40 ppm||0-400 & 0-2000 ppm|
|Call For Price||No|
|4-20 mA Output||No|
|Range||0-1ppm & 0-5ppm|